A Raphael: good day sir i have a fluke multimeter just as yours... when switching it on it will show f87 then show error and you cant do nothing else after that.... what do you think is the problem???
socksandtea: Excellent video and great explanations and demonstration. Thanks.
URanInTheUglyForest: Great video. After roughly 20 years, it's getting time to retire my 1st generation Fluke 87. Your video has explained the features well enough that I can see the new versus old improvements. Thanks!
aggienica: Hello, I hope someone can help me with this. At work we have a Fluke 189 and a 87V. Now, since I sometimes work with mV, I need 3 to 4 decimal points. Unfortunately, the Fluke 87V only has two decimal points. Can someone please tell me how I can add more decimal points so it can be like the 189.
Nicholas Aarons (Nick Mini): Awesome Video Dude. Keep up the great work. Nick.
THE MACKEM !!!: thank you..i didnt get mine for $99 haha....£299..but worth it..great video thanks again..
Alibek Alimbayev: I bouth mine from 2nd hand and it were CALibrated 3 times. Is it OK?
Alibek Alimbayev: Thanks! When i tested my MM F28IIEx firts value was correct 1.001 Second value a little scared me because it shows 0.3 constant, sometimes moved for a second to 0.4 Great explanation! Well done
Anthony Lopez: Great video. Hey do you think getting a 28II for $280.00 used in mint condition is a good deal? Not sure if I want to get a 28II or 85V. I currently am using a 117 so not real sure.
Jim Avila: Do you have one explaining the range function? I am measuring current with an AC clamp. When I set the Fluke to milliamp, I get a reading of 7.82 but the range is auto and shows 60 on the bottom. So is that 7.82milliamps or amps?
Clyde Str: you babble way to long !
SlyPearTree: I might be barking up the wrong tree since the video is so "old" but so is my meter. I have a second hand Fluke 87, not anything following the 87. I got it without a manual but I found one on line. Anyone know how to disable automatic power off on it? The method use in this video on the 87V does not work, I get no Poff display. The manual is not helpful unless the only way is with the MIN MAX option.
Brett West: Thanks Martin. That was very informative. I tried to check my fuses and it kept beeping "lead". Now I know how. Also,it was nice to learn how to check how many times it's been calibrated. Thanks for the help
Chingus696: Great job as always, Martin. I got a kick out of "POff... that's very polite. :)".
Charlie Pechiar: Great video!
JJ Lefty H: Thanks M8 I just bought a brand new 87-5 and I Love it!!! This vid was fantastic because like you said, Fluke doesn't have it imprinted on the DMM itself. I'm also going to watch the vid on Advanced measurements. TY Again, JJ......How do you flip the desk over in the vid? Which Prog are you using, .Camtasia?
Khaled Hariss: thx for the video, this is very helping, keep up the good work, I have this multimetter and it's great
georgecol19: My multimeter has the message ''CAL ERR'' on the screen. Do you know if I can solve it myself, and how? Thank you
Branislav Vojtek: i LIKE YOUR VIDEOS
Fluke 87 V / Fluke 87-5 / 28-II Power on options / features (Not obviously visible features)5
out of 5